Transcranial Magnetic Stimulator
- application: psychiatry, neurology
- solution for therapeutic repetitive transcranial magnetic stimulation (rTMS)
- number of pulses generated during one session – up to 10000
- Neuro-MS.NET software to control magnetic stimulator
Application: psychiatry, neurology
The principle of therapeutic magnetic stimulation is based on the following. The magnetic stimulator uses the magnetic pulses of short duration. The emerging high-intensity electromagnetic field easily penetrates through clothes, cranium bones and soft tissues. It impacts on deep nerve centers, peripheral nerves, cerebral and spinal cord, which is inaccessible for other kinds of stimulation.
This stimulation is commonly used to treat depressions (including severe and drug-resistant), tinnitus. Besides, there are positive and encouraging results obtained at rehabilitation of post-stroke patients and other neurological and psychiatric diseases.
Treatment of depressions, schizophrenia, obsessive-compulsive disorders, post-traumatic stress, manias and other psychiatric disorders.
Treatment of central and peripheral nervous system disorders: neuropathic pain, stroke, migraine, Parkinson disease, epilepsy, tinnitus etc.
Solution for therapeutic repetitive transcranial magnetic stimulation (rTMS)
It includes main unit with cooling system and cooled coil placed in a flexible arm that ensures fast and easy coil fixation in any position by rotating one screw. All that is mounted on a smooth-running trolley. The computer to control the stimulation can be also included in the delivery set. In this configuration system allows performing long-term sessions of biphasic rTMS with up to 30 Hz frequencies and up to 5 Hz maximum intensity.
Select any Neurosoft EMG recorder from a wide list of neurophysiological systems to record motor evoked potentials (MEPs).
Number of pulses generated during one session – up to 10 000
The cooling unit and cooled coils included in the delivery set of Neuro-MS/D are intended to avoid overheating of coils during the long-term treatment session. The cooling unit itself is a tank with the cooling liquid, the pump running the cooling liquid through the coil and the fridge. The unit allows increasing continuous operation of up to 10 000 pulses without overheating. Practically, it means that stimulator can operate for hours without overheating.
Neuro-MS.NET software to control magnetic stimulator
This delivery set includes a special Neuro-MS.NET software that allows customizing any stimulation protocols. This software keeps patient database, determines motor response threshold and controls the stimulation courses and sessions. You can perform the stimulation according to pre-defined ptorocols and also create your own ones or customize the preset stimulation programs.
Preset protocols are used to treat depression, Parkinson disease, stroke, tinnitus, etc.
|Magnetic stimulator "Neuro-MS/D", Therapeutic||1 pcs.|
|Neuro-MS/D main unit||1 pcs.|
|Neuro-MS/D cooling unit||1 pcs.|
|End cap||1 pcs.|
|Patient cap , size 42-54||10 pcs.|
|Patient cap , size 54-66||10 pcs.|
|Silicone oil , canister, 3 l||1 pcs.|
|Angulated figure-of-eight cooled coil 100 mm AFEC-02-100-C||1 pcs.|
|Hydraulic system adapter||2 pcs.|
|Cooling unit "Neuro-MS/D" contol cable||1 pcs.|
|Equipotential cable||1 pcs.|
|SCZ-1 mains supply cable||2 pcs.|
|USB cable , A-B||1 pcs.|
|K-3 flexible arm for coil positioning||1 pcs.|
|К-8 Coil holder , trolley/wall mounted||1 pcs.|
|T-4/A trolley||1 pcs.|
|NS4D 42006 3/8" quick connect coupling||1 pcs.|
|Technical Manual "Neuro-MS/D"||1 pcs.|
|Technical Manual «Coils for Magnetic Stimulators»||1 pcs.|
|User manual «Neuro-MS.NET », ver.2||1 pcs.|
|''Transcranial Magnetic Stimulation'' handbook by Moacyr Alexandro Rosa and Marina Odebrecht Rosa||1 pcs.|
|License for the use of software «Neuro-MS.NET»||1 pcs.|
|AFEC-02-100-C coil positioning tool||1 pcs.|
|Neurosoft equipment cover||1 pcs.|
The delivery set can differ from country to country. Request the actual delivery set for your country from your local representative.