Transcranial Magnetic Stimulator
- https://mycloudtms.com advanced solution for repetitive transcranial magnetic stimulation (rTMS)
- 20 Hz stimulation with 100% intensity
- number of pulses generated during one session – up to 10 000
- Neuro-MS.NET software to control magnetic stimulator
- application: psychiatry, neurology
For North America USA customers please visit this page.
*Get a quote, online demo or more information
Write to sales department
Advanced set for rTMS
Advanced therapeutic configuration of Neuro-MS/D covers all clinical needs of rTMS practitioners. Convenient system on a trolley includes high frequency and high intensity stimulator with effective cooling system and convenient flexible arm for coil positioning.
20 Hz stimulation with 100% intensity
Advanced therapeutic variant provides the possibility to increase the maximal effective stimulation frequency (the frequency when each stimulus in series has 100% intensity of motor threshold for most people) fourfold (up to 20 Hz) owing to extra power supply unit.
Number of pulses generated during one session – up to 10 000
The cooling unit and cooled coils are included in the delivery set of Neuro-MS/D to avoid overheating of coils during the long treatment session. The cooling unit itself is a tank with the cooling liquid, the pump running the cooling liquid through the coil and the fridge. The unit allows increasing continuous operation up to 10 000 pulses without overheating. Practically, it means that stimulator can operate for hours without overheating.
Neuro-MS.NET software to control magnetic stimulator
This delivery set includes the special Neuro-MS.NET software that allows to customize any stimulation templates. This software keeps patient database, detects motor response threshold and controls the stimulation courses and sessions. You can perform the stimulation according to already specified templates and also create own ones or edit the preset stimulation programs.
In advanced therapeutic configuration rapid stimulation protocols with up to 100Hz frequency as well as theta-burst stimulation (TMS) are available.
Preset templates for: depression, Parkinson’s disease, stroke, tinnitus etc.
Application: psychiatry, neurology
The principle of therapeutic magnetic stimulation is based on the following. The magnetic stimulator uses the magnetic pulses of short duration. The emerging high-intensity electromagnetic field easily penetrates through clothes, cranium bones and soft tissues. They exert influence upon deep nerve centers, peripheral nerves, cerebral and spinal cord, which are inaccessible for other kinds of stimulation.
This stimulation is commonly used to treat depressions (including severe and drug-resistant), tinnitus. Besides, there are positive and encouraging results obtained at rehabilitation of post-stroke patients and extensive list of other neurological and psychiatric diseases.
- Treatment of depressions, schizophrenia, obsessive-compulsive disorders, post-traumatic stress, manias and other psychiatric disorders.
- Treatment of central and peripheral nervous system disorders: neuropathic pain, stroke, migraine, Parkinson’s disease, epilepsy, tinnitus etc.
As a psychiatrist who does a lot of TMS treatments, I was interested when I found out that the NeuroSoft TMS system was FDA cleared. When I looked into it I found that it was the NeuroSoft TMS system which I was first introduced to in Singapore in 2015. I saw it demonstrated at every conference since that time but I did not appreciate it until I considered purchasing the system. The price is the most obvious attractive feature. It is not only about $10,000 less then the nearest competitor, but its performance matches and even exceeds some of the other TMS systems currently on the market. When my unit arrived and Fred showed me how it worked I was amazed at how simple it was. We can train our technicians on this machine in about a minute. The movement of the coil holder is smooth and fluid. This system runs very cool even when treatments are back to back for several hours. We have had no problem overheating. This NeuroSoft TMS system is a superb choice. We have seen equal results with it and have no complaints from patients or technicians. I can recommend this system highly
We bought a NeuroSoft system few months ago and this is our first TMS machine. We are very pleased with it so far. It gives us flexibility to perform wide range of treatments. No complaints from patients or technicians. Very easy to learn. We especially appreciate the smoothness of its work and the integration with the Neuro-MS.NET software. Many important treatment protocols for wide variety diagnoses are preconfigured in the software. As soon as you do a measurement the right algorithm takes place. You can do back to back treatments without machine being overheated. Synthetic oil cooling is a great solution! The machine is very quiet. The placement of a coil is very flexible, you have the freedom to choose the location very precise and perform focused and localized treatment. The price of the system is also very attractive compare to all other machines on the market. I would recommend this system to any TMS provider. We think this is the best TMS package out there as of yet
Neurosoft beats the competition in many ways, including ease of use, great interface, and equipped with software containing automated protocols. The team is outstanding with personalized service!
The NeuroSoft rTMS system is well built, easy to maneuver around the exam room, and is quieter than a formally used rTMS system. The software interface is very user friendly with nice nearly-automated patient treatment reports. We find it be a good value
Neuro-MS/D magnetic stimulator has 4 delivery set variants:
|List of Components||Variant 1||Variant 2||Variant 3||Variant 4|
|Neuro-MS/D main unit||1||1||1||1|
|Neuro-MS/D extra power supply unit||—||—||1||1|
|Neuro-MS/D cooling unit||—||1||1||1|
|Trolley with custors for magnetic stimulator||—||1||1||1|
|FISSO coil holder||—||—||—||1|
|Manfrotto coil holder||—||1||1||—|
|Big ring coil RC-02-150||1||—||—||1|
|Small ring coil RC-02-100||1||—||—||—|
|Figure-of-eight coil FEC-02-100||1||—||—||—|
|Cooled angulated figure-of-eight coil AFEC-02-100-C||—||1||1||1|
|HV cable for Neuro-MS/D extra power supply unit||—||—||1||1|
|Control cable for Neuro-MS/D cooling unit||—||1||1||—|
|Control cable for Neuro-MS/D cooling and expansion units||—||—||—||1|
|Touchscreen computer holder||—||1||1||1|
|Neuro-MEP-Micro (version 2009)||—||—||—||1|
|“Transcranial Magnetic Stimulation” handbook by Moacyr Alexandro Rosa and Marina Odebrecht Rosa||1||1||1||1|